« Schmitt-Cassegrain Collimation Made Easy Using a Duncan Mask
Correctly position the Duncan Mask so each collimation screw is opposite a 60 degree window as indicated by the arrow
Your email address will not be published. Required fields are marked *
Save my name, email, and website in this browser for the next time I comment.
Current ye@r *
Leave this field empty
This site uses Akismet to reduce spam. Learn how your comment data is processed.